Voltmeters – Types & Their Descriptions Voltmeter What is a Voltmeter? A voltmeter is an instrument used to measure voltage or electrical potential difference between two points in basic electric circuits. Analog voltmeters move a pointer across a scale in proportional to the voltage of the circuit. Voltmeters may have an accuracy of a few percent of full scale, and are used with voltages from a fraction of a volt to several thousand volts. Two common voltage measurements are direct current (DC) and alternating current (AC). Although voltage measurements are the simplest of the different types of analog measurements, they present unique challenges due to noise considerations. Analog voltmeters move a pointer across a scale in proportion to the voltage of the circuit; digital voltmeters give a numerical display of voltage by use of a converter. Different types of voltmeters are 1. Analog voltmeters 2. VTVMs and FET VMs 3. Digital voltmeters 1. Analog Voltmeters Analog voltmeter includes deflecting type of indicating voltmeters. These are of like moving iron, moving coil, electrostatic types of voltmeters. Moving coil instruments are of two types namely permanent magnet and dynamo meter types Moving-coil instruments with a permanent-magnet field respond only to direct current. Moving coil instruments consists of permanent magnet to produce magnetic field, and a coil which is wounded on soft iron piece and rotates about its own vertical axis. When the current starts flowing through this coil, deflecting torque is produced according to Lorenz force equation. This torque is directly proportional to the voltage across that particular circuit. A DC voltmeter is constructed by connecting the resistor in series this instrument and also a very high resistor in parallel with the circuit where we want to measure the voltage. Dynamo meter type Moving coil instrument consists of two coils one is fixed and other is rotating. The interaction of the two fields produced by the pair of fixed coil and moving coil produce a deflecting torque. These are used in DC measurement circuits only this makes this instrument to less usage. Moving-Coil Voltmeter Advantages of Moving Coil Instruments Scale is uniform Easily extended for multi range measurements. Low power consumption Stray load currents are very small as compared to moving iron instruments. Moving Iron Voltmeter While moving iron instruments are used in AC circuits. Electromagnetic instruments are divided into simple moving iron, dynamo meter type and induction types of instruments. Again moving iron classified as attraction and repulsion types of instruments. In this also consists of soft iron which is of movable and fixed coils. The interaction of fluxes produced by these two elements produces the deflecting torque. The ranges of these instruments are extended by keeping the resistors in series with the coil. Some of the disadvantages are non uniform scale, stray field current effects on the instrument etc. Advantages of moving iron instruments These are used for both AC and DC measurements. Low cost compared to moving iron instruments. Torque to weight ratio is high Electrostatic Voltmeter Electrostatic Voltmeters operating on the electrostatic principle use the mutual repulsion between two charged plates to deflect a pointer attached to a spring. These types of instruments are used for high voltage AC measurements as well as DC. These are of electrostatic disc type capacitor connected across the circuit which is to be measured. The electrostatic voltmeters can be categorized into three types based on the mechanical configuration. Those are repulsion, attraction, and symmetrical. Deflecting system consists of deflector which is suspended from a torsion filament or It can be pivoted by the bearings. Arrangements the elements in this type of instrument mad with some special elements like capacitative elements, including parallel plates, concentric cylinders, hinged plates, etc Motion Damping torque is provided by air or liquid damping vanes or by eddy current damping. Advantages of Electrostatic instruments These only draw the currents at DC is leakage current and current required to charge the capacitive elements High sensitivity Capable of measuring smallest charge voltages High range of voltage measurement possibility nearly 200KV 2. VTVMs and FET-VMs Vacuum Tube Volt Meter (VTVM) These types of instruments handle DC voltage, AC voltage, and resistance measurements. In this type of voltage measuring device electronic amplifier is used in between the input and the meter. Due to this arrangement the current drawn from the circuit under test is reduced. The range of resistances used at the input side in range of 1-20 mega ohms. By these resistances variation we can select the range to be measured. If this instrument uses the vacuum tube in the amplifier then it is called as vacuum tube voltmeter. These are used in high power AC measurements. As the invention of solid state devices used in the amplifiers, these type of voltmeters are called FET-VM. Advantages These have high input impedance hence loading error is less Non linearity’s is almost eliminated Ability to indicate slowly varying voltages. 3. Digital Voltmeters Digital Voltmeters Voltmeter accuracy is affected by many factors, including temperature and supply voltage variations. DVMs display the measured voltage using LCDs or LEDs to display the result in a floating point format. Obviously, if voltage measurements are taken and the results are displayed digitally with LED or LCD displays, the instrument has to contain an A/D converter. Using a programmed micro controller, ADC and LCD display, the following circuit is ready to provide accurate digital display of analog values from 0 to 15 volts DC. These are used because of the properties like accuracy, durability and extra features. These are completely eliminates the parallax errors. It will convert the signal under test and then amplifies it. Advantages of Digital Voltmeters Reduces the parallax errors Auto ranging Automated Polarity High resolution instrument includes high precision. Digital Electronic Voltmeter Circuit Diagram Digital Voltmeter Electronic Circuit Diagram The digital voltmeter design uses a microcontroller which is said to be highly efficient in handling the data carrier operation in terms of being faster, error-free and accurate. Rather than using the absolute analog ways of finding out the voltages, the digital voltmeter provides much more precise and accurate values of voltages in a given circuit in the range of the voltmeter. Watch the following video to understand the functionality of Digital Voltmeter: You may get good knowledge on electronic circuits and various ideas on projects of electronics by visiting this blog regularly. You may subscribe to this blog for regular updates. Photo Credits: Voltmeter by solidswiki Moving-Coil Voltmeter by wikimedia Moving Iron Voltmeter by electrical4u Electrostatic Voltmeter by ayunor Vacuum Tube Volt Meter (VTVM) by ohio Digital Voltmeters by imimg Share This Post: Facebook Twitter Google+ LinkedIn Pinterest Post navigation ‹ Previous Transistors – Basics, Types & Baising ModesNext › RS232 – Basics, Applications and Interfacing Related Content Semiconductor Fuse : Construction, HSN code, Working & Its Applications Displacement Transducer : Circuit, Types, Working & Its Applications Photodetector : Circuit, Working, Types & Its Applications Portable Media Player : Circuit, Working, Wiring & Its Applications Comments are closed.